Display : 4 digits, LCD display.
Range : Moisture Content: 0-50%
Temperature : 16-65 ºC
Accuracy : Moisture Content: ± (0.5%)
Temperature: ± 0.9 ºC
2. Measuring Procedure:-
(i) Hold the handle measurement needles and leave needles
in the air, i.e. don’t let needles touch anything except air.
(ii) Set the Function Switch to the Moisture Content (H2O)
position to switch on the power supply.
(iii) You will hear 2 times of “click” sound. That is self-calibration by the instrument.
(iv) Insert two steel measurement needles into the paper or box in correct way to be measured. The reading on the Display is the measurement value.
(v) Hold : To keep the maximum value, just push the Hold button for a while,
the symbol “max” will appear on the display. To cancel the hold value,
just push Hold button again for a while when the symbol “max” disappears.
Payment Terms : L/C, T/T, Western Union
Packaging Details : Wooden Packing
316, Avas Vikas Colony, Roorkee, Uttarakhand - 247667, IndiaView Mobile No.